The Nexus menu bar contains the following controls:
The File menu contains the following options:
- Save (shortcut CTRL+S) Saves the current data for subjects enabled in the Subjects Resources pane to the .c3d file for the current trial.
Copy As Displays the Choose a c3d file dialog box in which you can enter or select the name of the file to copy to. Makes a copy of the current trial .c3d, .vsk, and .mp files. Using this feature, you can archive copies of a .c3d file to document its processing progression.
You are prompted to save the copy in the same location in the Data Management hierarchy as the original file, and after saving it will be available in the Data Management pane.
- Import XCP Displays the Choose an XCP file dialog box in which you can select the desired Vicon camera calibration parameters (.xcp) file to load in Nexus. Use this command to import .xcp files created in Vicon Nexus.
- Export XCP Displays the Choose an XCP file dialog box in which you can specify a calibration parameters (.xcp) file into which to export the current Vicon camera settings. Use this command to export .xcp files created in Vicon Nexus.
- Exit Closes the Vicon Nexus application window. If you have not saved any changes made to trial data, Nexus displays a prompt to enable you to save changes to trial data, subjects, or both, before it closes.
The Edit menu contains the following options, which take effect only when the system is in Pause or Offline mode:
- Undo (shortcut CTRL+Z) Undoes the last action. This command is available only after a relevant action has been performed. The name of the Undo command changes to reflect the latest action, for example, Undo Import XCP. You also can hover the mouse pointer over the Undo button to display a tooltip that identifies the action to be undone. Some actions, such as reconstructing data, cannot be undone.
- Redo (shortcut CTRL+Y) Reinstates the previously undone action. This command is available only after an Undo command has been performed. The name of the Redo command changes to reflect the latest action, for example, Redo Import XCP. Hover the mouse pointer over the Redo button to display a tooltip that identifies the action to be redone. Any action that was undone can be redone.
The Window menu contains the following options:
Simple Capture Mode (shortcut CTRL+H). Displays a window containing only the minimum of controls necessary for capture, facilitating the capture of multiple trials. For more information, see.
- New floating workspace. Opens a separate floating view pane. For information, see View pane.
- Reset to default layout Resets the Nexus panes to their layout when Nexus was installed.
- Toolbar Opens the Customize Toolbar dialog box, in which you can add, remove, and customize buttons on the Nexus toolbar. For more information, see Customize Toolbar dialog box.
Quick Reports (shortcut F4) Opens a Quick Reports window, in which you can display multiple graphs of model outputs. For more information, see.
- Sounds (shortcut F6) Opens the Sounds dialog box, in which you can view or change the sounds that are used to alert you to Nexus events. You can also disable the sounds individually or disable all sounds. For more information, see Sounds dialog box.
- Options (shortcut F7) Opens the Options dialog box, in which you control the way data is displayed. For more information, see .
- Plug-ins Displays the Plugins dialog box in which you can view and manage plug-in modules that have been loaded in Vicon Nexus.
- Manage Control Authorizations Opens the Vicon Control dialog box, where you can authorize or revoke authorizations for instances of Vicon Control that are connected to the Vicon system. See Vicon Control dialog box.
- Footstrike Monitor Settings Lets you set options for foot strike monitoring. See Automatically assess foot strikes in the Vicon Nexus User Guide.
- Close Communications Pane on Trial Load Enables you to choose whether to show or hide the Communications pane when you load a trial.
If selected, the Communications pane is minimized at the bottom of the Nexus window the next time you load a trial. The Communications pane remains hidden until you double-click one of the tabs at the bottom of the Communications pane. Your chosen setting for this option is retained when you exit and restart Nexus. It can be set as required by each Nexus user.
The Nexus Quick Reports window enables you to display multiple graphs of model outputs normalized over the gait cycle (like a simplified Vicon Polygon graph view).
To display model outputs in Nexus Quick Reports:
- On the Window menu, click Quick Reports, or press F4 to open the Nexus Quick Reports window.
- In the main Nexus window, ensure that the trial (*.C3D file) that you want add to Quick Reports is the current trial.
- From the Pipeline Tools pane, expand Data Processing and double-click the pipeline operation, Add Trial to Quick Report to add it to the current pipeline.
- In the Current Pipeline list, right-click the Add Trial to Quick Report operation and click Run select Op.
For each trial that you want to add, repeat step 2–4.
To add trials to Quick Reports, you can also:
- Drag and drop one or more trials (*.C3D files) from Windows Explorer into the Quick Reports window; or
- Click the Add To Quick Report button on the Nexus toolbar (adds the current trial).
In the Nexus Quick Reports window, the added trials are displayed in the Trials list, by default on the right of the window.
Ensure the required cycle is selected in the Left and Right columns.
To change the options for all the trials, click the downward arrow to expand the Trials section and click the appropriate option at the top of the columns.
To display model outputs in the graphs, in the Model Outputs list, expand the relevant node and then select the required check box(es).
The selected outputs are displayed as graphs in the Nexus Quick Reports window.
By default a red line represents data from the left side and a green line represents data from the right. The trial currently selected in the Report Options list is represented by a heavier line on the graphs. To change green and red to solid and dashed lines, in the Options dialog box (press F7), on the left, click Quick Reports and then clear Use Context Colors. To change the colors displayed for each trace on the graph, click the color(s) that you want to change.
Expand the Gait Cycle Analysis section at the bottom left of the window and ensure the Distance Markers and Units are as required.
By default, the Parameter column displays average values across all selected trials. To view the average value for each trial, expand the relevant parameter.
- To export a PDF of the data in the Nexus Quick Reports window:
- Expand the Export Options section and clear or select the required options.
- From the Title Alignment list, select the required option.
- Click Export and in the Export to PDF dialog box, enter or browse to the required location.
- Click Save.
- To save your work, click the Save button at the bottom of the Report Options pane and select the appropriate option:
- *. qrd (Quick Reports Document): Saves the current Quick Report. Select this option if you have to suspend work on the report and want to carry on where you left off later.
- *. qrt (Quick Reports Template): Saves the layout of the Nexus Quick Reports window (ie the options currently selected in the Quick Reports window). Select this option if you want to produce reports of the same type of information for later trials. Quick Reports templates that are saved to the default location ( C:\Users\Public\Documents\Vicon\Nexus2.x\Configurations\QuickReports ) are listed when you click the Apply Report Template button .
About the Nexus Quick Reports window
The Nexus Quick Reports window comprises two sections:
- Controls are contained in a dockable Report Options pane, which by default is displayed to the right of the window. Expand the Trials, Model Outputs and Export Options sections by clicking the downward arrow to the right of each section divider. To move the controls pane, drag the Report Options title. The pane can be floated or docked to the left or right.
- Graphs of the data selected in the Report Options pane are shown in the main workspace. The current subject’s name and model template are shown as a title at the top.
To the bottom left of the main workspace, Gait Cycle Analysis parameters for the loaded trials are displayed.
In the large text box on the right, you can enter a description or notes, etc, which can be formatted using the buttons on the toolbar above the text box.
Report Options controls
In the Report Options pane, the following controls are available:
Enables you to select the subject whose data you want to view.
Lists the trials that you have added to the report. In this list, you can select or clear the relevant check boxes to display the required trial(s). To show/hide all trials, expand the Trials section and select or clear the check box at the top. The outputs for selected trials are displayed in each graph.
The Cycle dropdown lists at the top and in the Left and Right columns of the Trials list enable you to select which gait cycles to show for the left and right contexts for each trial.
To select trials, click (or Shift+click/Ctrl+click) in the Trial column of the required row(s) in the list. The selected row(s) in the list are highlighted with a blue background and by thicker lines in the graph.
To remove selected trials from the report or to mark selected trials in Data Management (eg, for use in Vicon Polygon), click Remove Selected or Mark Selected at the bottom of the list.
Model Outputs list
Displays available model outputs from the trials that have been added to the report.
|Export button||Opens the Export to PDF dialog box, which enables you to save the contents of the current Nexus Quick Reports window to a *.PDF file, with a name and location that you specify.|
Enables you to save the current Quick Report either as a document (*.qrd) or a template (*.qrt).
Enables you to enter or browse to a Quick Report document (*.qrd) or a template (*.qrt).
Apply Report Template button
|Enables you to apply a report template that was saved to the default location (C:\Users\Public\Documents\Vicon\Nexus2.x\Configurations\QuickReports).|
Sounds dialog box
Nexus can play sounds through your PC's speakers to alert you when a Nexus event has taken place (for example, operation completion, system status, and issues). This keeps you informed of system status while you're still in the volume, so that you don't have to spend time returning to the PC to look at the screen.
You can access this dialog box from the Window menu or by pressing F6.
By default, Nexus uses speech sounds to alert you to the following events:
- Calibration start
- Wand Wave complete
- Calibration complete
- Calibration failed
- Origin set
- Capture started
- Capture ended
- Capture failed
- Camera bumped
- Pipeline ended
- Pipeline failed
Nexus is supplied with a set of default sound files (.wavs). You can modify the sounds that are used for each event and you can disable sounds individually or disable all sounds.
To disable all sounds, in the Sounds section, clear the Enabled check box.
To disable one or more sounds, click the relevant drop-down arrow and select (None) from the list.
To choose one of the other sounds supplied with Nexus, click the drop-down arrow and select the required sound from the list.
To substitute your own sounds for those supplied with Nexus, click the relevant ellipsis (...) and enter or browse to the location of the required .wav files.
Options dialog box
The Options dialog box enables you to control the way data is visualized in the view panes . You can access this dialog box from the Window menu or by pressing F7.
The Options dialog box includes a configuration area:
This enables you to save any changes you make in the Options dialog box to a configuration file, with the extension .Options. You can then re-use your saved configuration file as required, for example, you could save a different set of options for each type of motion capture application that you use.
In addition to creating customized Options configurations, the Configuration menu button enables you to rename, import, reload and delete configurations, and refresh the list.
To configure settings in the Options dialog box:
- Open the Options dialog box in either of these ways:
- Click the Window menu and then click Options; or
- Press F7.
- In the list of options on the left side of the dialog box, click an option whose properties you wish to view or change. You can now:
- Select the check box to switch on the functionality or clear the check box to switch off the functionality for the option.
- In the Properties section on the right side of the dialog box, view or change settings as desired for any available properties. (To see additional settings that may be available for an option, click Show Advanced.)
- Repeat step 2 for each property whose settings you wish to configure.
- To save your settings, do one of the following:
- To save your settings as the current configuration and close the Options dialog box, click Close; or
- To give your configuration a name, so that you can easily find it for re-use later, click the Configuration menu button, then click Save As, enter a name in the dialog box, click OK, and then choose whether to save your configuration as Shared or Private.
Vicon Control dialog box
The Vicon Control dialog box enables you to carry out the following operations:
- To grant unprompted access in future, ensure the relevant device's check box is selected in the Known Devices list.
In future, the device will be able to connect without having to be re-authorized.
- To permanently revoke access, clear the device's check box in the Known Devices list.
The device is disconnected and in future, it will not be able to connect. This is useful if you accidentally authorized a device, or if you need to remove an authorized device from the system, for example, if it is lost or sold.
- To remove a connected device from the Known Devices list and force re-authorization on the next attempt to connect, select the device and then click Forget Device.
To reconnect, the device will have to send an authorization request and be re-authorized.
- To remove all connected devices from the Known Devices list and force re-authorization on the next attempt to connect by any of the listed devices, click Forget All.
The Help menu contains the following commands:
- View latest help from vicon.com Displays the most recent version of the live online help.
- View installed help Displays the help file (PDF) that was installed with Vicon Nexus.
- Check for updates Detects whether any Nexus updates are available.
- About Vicon Nexus Displays the Vicon Nexus startup screen, in which you can view version and license server information about the installed release of Nexus.
- Hotkeys Displays a list of Nexus shortcuts and hot keys.
- Vicon Product Licensing Displays the Vicon Automated Unified Licensing Tool (VAULT), which enables you to manage your Vicon product licensing.