You manage the connection and configuration settings for supported analog devices included in your Vicon system architecture with the appropriate Devices sub-node on the System tab of the Resources pane.
For information on force plates, see Force plate nodes.
Analog device node context menu
You can select the following options from the context menu displayed when you right-click on a sub-node for an analog accelerometer or EMG:
Calibrates the force plate. This process eliminates any significant offset between the force plate's nominal output levels at rest and its theoretical zero level. Select this option if, after setting the force plates to the electrical zero level, small differences remain between the theoretical zero level and the observed output level. This option is also available in the General Properties pane.
Remove the device entry from the System resource tree.
You can select from a list of available outputs from the context menu that is displayed when you right-click on a Generic Analog sub-node. Nodes representing the outputs you select are added below the Generic Analog sub-node in the System Resources tree.
Analog EMG device node properties
You can configure settings in the following sections in the Properties pane for analog EMG devices:
A user-defined display name for the device. Default: Blank
Delay Compensation (s)
The delay compensation value (in seconds).
Amplifier Gain (V)
The voltage gain scale factor. Can be set between 1 - 1000. Default: Depends on device
The Vicon connectivity device to which the physical force plates device is connected.